Find the most up-to-date version of ISO at Engineering Intravascular catheters – Sterile and single-use catheters – Part 3: Central venous catheters (ISO ); German version EN ISO. ISO Intravascular catheters — Sterile and single-use catheters — Part 1: General requirements. ISO Intravascular catheters — Sterile and.
|Published (Last):||9 June 2015|
|PDF File Size:||20.15 Mb|
|ePub File Size:||6.87 Mb|
|Price:||Free* [*Free Regsitration Required]|
Disclosure io Promote the Right To Information Whereas 10555- Parliament of India has set out to provide a practical regime of right to information for citizens to secure access to information under the control of public authorities, in order to promote transparency and accountability in the working of every public authority, and whereas the attached publication of the Bureau of Indian Standards is of particular interest to the public, particularly disadvantaged communities and those engaged in iao pursuit of education and knowledge, the attached public safety standard is made available to promote the timely dissemination of this information in an accurate manner to the public.
The number of significant places retained in the rounded off value should be the same as that of the specified value in this standard.
ISO – Sterile, single-use intravascular catheters — Part 3: Central venous catheters
Attention is particularly drawn to the following: NOTE 3 At the time of publication of this part of ISOthere is no acceptable, validated test method to determine radio-detectability. This does not preclude the free use, in the course of implementing the standard, of necessary details, such as symbols and sizes, type or grade designations.
Technical Corrigendum 1 to the above International Standard has been given at the end of this standard.
From the first mark, the distance between marks shall not exceed 5 cm. Bureau of Indian Standards BIS is a statutory institution establislied under the Bureau of Indian Standards Act, to promote jiarmonious development of the activities of iao, marking and quality certification of goods and attending to connected matters in the country. Standards are also reviewed periodically; a standard alongwith amendments is reaffirmed when such review indicates that no changes are needed; if the review indicates that changes isso needed, it is taken up for revision.
BS EN ISO 10555-3:2013
Certain conventions are, however, not identical to those used in Indian Standards. An example of a suitable apparatus is shown in fig- ure A. Until that time, a manufacturer may label his product “radio-opaque” provided he can support this claim by demonstrating that he has an appropriate method for showing radio-opacity.
NOTE 4 it is recommended that the distance marks be 1 cm apart on that portion of the catheter likely to be of im- portance to the user in positioning the catheter and moni- toring catheter migration. The following standards contain provisions which, through reference in this text, constitute provisions of this part of ISO Search the history of over billion web pages on the Internet. Round the calculated average water flowrate to the nearest whole number of millilitres.
Users of Indian Standards should ascertain that they are in possession of the latest amendments or edition by referring to the latest issue of ‘BIS Catalogue’ and ‘Standards: An approved test method for producing a value of radio-detectability will be established.
The minimum force at break of all other parts of such catheters shall comply with ISO 1: All standards are subject to 10555-3, and parties to agreements based on this part of ISO are encouraged to investigate the possibility of applying the most recent editions of the standards indicated below. The corresponding Indian Standards which are to be substituted in their places are listed below along with their degree of equivalence for the editions indicated: In this adopted standard, reference appears to certain International Standards for which Indian Standards also exist.
International Standard ISO Review of Indian Standards Amendments are issued to standards as the need arises on the io of comments. Copyright BIS has the copyright of all its publications.
For the purpose of deciding whether a particular requirement of this 10555- is complied with, the final value, observed or calculated expressing the result of a test or analysis, shall be rounded off in accordance with IS 2: Details of which is given in National Annex A.
General requirements IS01 The details of conditions under which the licence for the use of the Standard Mark may be granted to manufacturers or producers may be obtained from the Bureau of Indian Standards.
This Indian Standard has been developed from Doc No.: Table 1 — Minimum force at break of soft tips of lengtli not exceeding 20 mm 4. No part of the these publications may be reproduced in any form without the prior permission in writing of BIS. At the time of publication, the editions indicated were valid.